 Research Article
 Open Access
 Published:
Data Fusion for Improved Respiration Rate Estimation
EURASIP Journal on Advances in Signal Processing volumeÂ 2010, ArticleÂ number:Â 926305 (2010)
Abstract
We present an application of a modified KalmanFilter (KF) framework for data fusion to the estimation of respiratory rate from multiple physiological sources which is robust to background noise. A novel index of the underlying signal quality of respiratory signals is presented and then used to modify the noise covariance matrix of the KF which discounts the effect of noisy data. The signal quality index, together with the KF innovation sequence, is also used to weight multiple independent estimates of the respiratory rate from independent KFs. The approach is evaluated both on a realistic artificial ECG model (with real additive noise) and on real data taken from 30 subjects with overnight polysomnograms, containing ECG, respiration, and peripheral tonometry waveforms from which respiration rates were estimated. Results indicate that our automated voting system can outperform any individual respiration rate estimation technique at all levels of noise and respiration rates exhibited in our data. We also demonstrate that even the addition of a noisier extra signal leads to an improved estimate using our framework. Moreover, our simulations demonstrate that different ECG respiration extraction techniques have different error profiles with respect to the respiration rate, and therefore a respiration raterelated modification of any fusion algorithm may be appropriate.
1. Introduction
Estimation of respiratory rate from waveforms recorded from passively breathing subjects is notoriously difficult, due in part to the nonstationary nature of the signals and in part the frequent nonstationary noise [1]. Methods for recoding a time series of respiratory effort include impedance pneumography (differential changes in capacitance recorded at high frequencies), impedance plethysmograpy (stretch sensors on the chest wall), and flow thermography which measures the changes in temperature of air flow as it moves in and out of the mouth and/or nose over a thermistor. Of these methods, the impedance pneumogram (IP) is the most common method employed in hospitals [2, 3], where a current is passed between two ECG electrodes and the differential change in capacitance due to air volume changes is measured. It is also possible to automatically record respiratory activity from accelerometers, laserreflectivity, ultrasound, or by audio or video processing. However, such signals are not commonly recorded in most patient monitoring scenarios.
Another class of respiratory signal sources comes from measurement of indirect effects on cardiovascular physiology. Respiratory information is present in other commonly monitored physiological signals, such as the electrocardiogram (ECG) [4], photoplethysmogram (PPG) [5], arterial blood pressure (ABP) [6], and the peripheral arterial tonometry (PAT) waveforms.
Amplitudebased ECGderived respiration (EDR) algorithms have been reported to perform satisfactorily when only singlelead ECGs are available, as is usually the case in sleep apnea monitoring. When multilead ECGs are available, EDR algorithms based on either multilead QRS area or QRSVCG loop alignment are preferable [7]. The reason is that due to thorax anisotropy and its intersubject variability together with the intersubject electrical axis variability, respiration influences ECG leads in different ways; the direction of the electrical axis, containing multilead information, is likely to better reflect the effect of respiration than wave amplitudes of a single lead.
Although the IP waveform is usually more representative of the air flow (but not volume), the IP is often unusable for accurate respiration rate evaluation around 37% to 61% of the time [8, 9].
Literature on deriving a respiratory signal from other related signals is dense in the case of the electrocardiogram, but relatively sparse for other signals (such as the photoplethysmogram and blood pressure waveform [10]). The field of respiration rate estimation from respiratory signals (whether derived or not) is scantily covered in the public literature, particularly with respect to large data sets. Key works by O'Brien and Heneghan [11], de Chazal et al. [12], Ishijima [13], Park et al. [14], and Tarassenko et al. [1] highlight the importance of modeling noise and combining information from multiple ECG leads and sensor modalities to compensate for noisy measurements. To our knowledge, very little work has been published in the domain of respiration signal quality estimation.
The approach detailed in this paper is based on our earlier work related in the context of robust heart rate and blood pressure estimation [15, 16]. Figure 1 summarizes the proposed robust respiration rate estimation technique using a signal fusion framework and signal quality indices. Fusion is the processes of combining signals from multiple instruments and sources in order to reduce measurement noise and improve overall signal quality. The field of data fusion in the context of physiological signals is described elsewhere [15]. When applied to respiration in particular, the key works of Mason [17] and Mason and Tarassenko [18] form the basis for our approach. However, our key innovation is the inclusion of a signal quality metric (as well as the past behavior of each signal) to control the KF noise covariance estimate and decide automatically how to weight each source of information. This is in contrast to standard industry approaches such as the work of Park et al. [14], who developed a system for deciding which single channel of ECGderived respiration was the most informative. In this article we present a method for combining all the available information from every channel, even if it is noisy, to produce a superior estimate.
Given that artifacts in the ECG, IP, and PPG or PAT do not always manifest simultaneously, it is likely that fusing respiratory rate data from each signal will result in an overall improved estimate of this parameter [17, 18]. The difficulty lies in determining which signal to trust, for if we add parameter estimates from noisy signals to those from clean signals, we may in fact degrade any one single parameter estimate. We present a solution to this problem for accurate robust estimation of the respiration rate using signal quality indices (SQIs) and a modified Kalman Filter (KF) fusion framework which uses the SQIs to adaptively update the KF noise covariance estimate. The SQIs are derived in real time and therefore no assumptions concerning the signaltonoise ratio (SNR) are required.
The paper is organized as follows Section 2.1 describes the datasets used in this study. Section 2.2 introduces the methods for deriving respiratory waveforms from the ECG and PAT signals, Section 2.3 provides an overview of a previously introduced ECG signal quality index and expounds on the applicability of a newly developed signal quality measure to assess the quality of the derived respiratory waveforms, and Section 2.4 describes the utilized respiratory rate extraction algorithm, followed by a discussion of the proposed KFbased fusion framework in Section 2.5. In Sections 3.1 and 3.2 we present the results of applying the proposed respiratory rate estimation algorithm to simulated data and real recordings from 30 subjects during an overnight polysomnographic study.
2. Material and Methods
2.1. Simulations and Data Collection
Two datasets were used for the analysis: one set of computersimulated ECG recordings with known respiratory signal modulation, and a set of real recordings from 30 subjects during an overnight polysomnographic study.
2.1.1. Simulated Data
Our simulated data were based on the synthetic ECG generation framework described by Clifford et al. [19], where we presented generalizations of our previously published artificial models for generating multichannel ECG to provide simulations of cardiac rhythms. Using a threedimensional vectorcardiogram (VCG) formulation, we generated the normal cardiac dipole for a patient using a sum of Gaussian kernels, fitted to real VCG recordings. The RR interval time series were generated using our previously described model whereby time and frequencydomain heart rate (HR) and heart rate variability (HRV) characteristics could be specified. Furthermore, following AstrÃ¶m et al. [20] we incorporated a model of respiratory sinus arrhythmia (RSA) and RSamplitude modulation to reflect influence of respiration on ECG. All the ECG signals were generated with a sampling frequency of 500â€‰Hz and 16bit amplitude resolution. Finally, realistic noise consisting of a combination of white noise (with SNRs of 10, 20, and 40â€‰dB), baseline wander, muscle artifacts, and electrode motion (obtained from the MITBIH Noise Stress Test Database [21]) were separately generated and added to the simulated ECG. This simulation study has been used to compare breathing rate estimates from individual respiratory waveforms as well as the combined breathing rate estimates using the proposed KFbased fusion framework.
2.1.2. Real Data: Overnight Polysomnography
To evaluate our methods on real data, we chose a database of 30 subjects undergoing overnight polysomnogram analysis. Recorded signals included one lead of ECG (V6), 4 channels of respiratory waveform data (chest and abdomen plethysmograph, nasal and oral thermistor, and nasal pressure), and one channel of PAT (Itamar Medical, Israel). The PAT signal is a pulsatile waveform recorded at the peripheral artery (fingertip), which is much like the ABP or PPG in morphological appearance, and reflects the rapid changes in blood pressure at the periphery from beat to beat. In particular, the PAT waveform, rather like the ABP or PPG waveform, exhibits amplitude fluctuations due to respiration.
All channels of data were recorded at 500â€‰Hz, 16 bits, with the exception of the PAT which was recorded at 100â€‰Hz. The length of each recording varied between 6 hours and 8 hours. Subjects were enrolled in the study for screening for sleep apnea, with an apneahypopnea index (AHI) ranging from 0 to 69.3 events/hour with a mean AHI of 14.0 events/hour. Since the respiration rate was not scored by humans, no gold standard measure of respiration rate was available. To provide a gold standard, we used the highest quality signal (the nasal and oral thermistor) and used the same AR spectral estimation technique for rate estimation and the signal purity for noisy section rejection.
2.2. Deriving Respiratory Waveforms
ECG beat detection was performed using a combination of two open source QRS detectors: Hamilton and Tompkins' eplimited QRS detector [22] and Zong's wqrs algorithm [23]. Beat detection for the PAT waveform was achieved using Zong's wabp blood pressure onset detector [24]. We considered three categories of derived respiratory waveforms.
(1) ECGDerived Respiration (EDR). We employed two forms of EDR: QRS area summation (EDR_G) and RS amplitude tracking (EDR_RS) [4].
(2) RSADerived Respiratory Waveform. Respiratory Sinus Arrhythmia (EDR_RSA) is known to contain a strong respiratory component, amongst other components [4].
(3) PATDerived Respiratory Waveform. The pulse amplitude is known to be modulated by respiration. Therefore, if we pick the onset and peak of each pulse (using an open source algorithm wabp [24]), we may derive an oscillatory signal indicative of the respiration effort.
Since normal range of breathing rates in adult humans is between zero (apnea) and 60 breath/minute (extreme hyperventilation), each respiratory waveform was resampled to 4â€‰Hz using a cubic spline method. Each respiratory waveform was segmented into 20â€‰s windows with 15â€‰s overlap from which the breathing rates were estimatedâ€”therefore, after an initial delay of 20â€‰s the breathing rate estimates were updated every 5â€‰s. These estimated values of the breathing rate were the measurement inputs to the Kalman filter.
2.3. Signal Quality Metrics
We have described our approach to determining the quality of the ECG previously [15, 16]. Briefly, we combine measures of abnormal statistics and power spectral density distribution with measures of QRSdetection mismatches to provide an overall quality estimate (between 0 being poor and 1 being excellent) for any given segment of ECG. The estimation of PPG signal quality is described in Gil et al. [25, 26]. We have extended these ideas to PAT and respiratory signal quality. Fidelity of respiration rate extraction is directly related to the periodicity of respiration waveform. A regular breathing pattern produces a highly sinusoidal reparation waveform with the dominant frequency at the frequency of breathing rate. A power spectralbased respiration rate extraction can be accomplished by identifying the most prominent peaks in any of the respiration waveforms (PATderived respiration, or ECGderived respiration waveform). Each peak is characterized by its frequency, amplitude, or its relative coherence with respect to similar spectral peaks in other respiratory waveforms. One measure of characterizing spectral characteristics of a signal is through Hjorth descriptors [26]. The n thorder spectral moment is defined as
where is the power spectrum of the signal as a function of angular frequency: , with being in units of cycles/second. A particularly useful descriptor in the context of estimating the dominant frequency and assessing the quality of a signal with periodic components (such as respiratory waveform) is the socalled spectral purity waveform and is defined as [27]
Here the term "purity" refers to the presence of a single signal frequency, as we would expect in an ideal respiratory waveform. In the case of a periodic signal with a single dominant frequency, takes the value of one and approaches zero for nonsinusoidal noisy signals. One of the attractive features of Hjorth descriptors is the feasibility of their calculation in time domain with low computational cost.
2.4. Deriving Respiratory Rates
The breathing rate extraction method was based on the work of Mason and Tarassenko [18], who utilized autoregressive (AR) modeling, a parametric spectral analysis technique. One advantage of AR modeling of spectral analysis over the traditional Fourier transformbased methods is its superior performance when the number of available data points is small (100 points). The steps involved in extracting breathing rates from the respiratory waveforms are as follow.

(i)
Fit an allpole model to each 20?s segment (use Akiake Criteria [28] to decide on the right model order).

(ii)
Pick an accepted breathing range:

(a)
lower: 4 breath/minute (0.066?Hz or 12?deg).

(b)
upper: 55 breath/minute (0.917?Hz or 165?deg).

(a)

(iii)
Exclude those poles not within the range.

(iv
)Keep all the poles with magnitude of at least 95% of the highest magnitude pole.

(v)
Pick the pole with the smallest angle. The frequency associated with this pole is the breathing frequency.
The window was advanced by 5â€‰s and the above process repeated so that a respiration rate was available every 5â€‰s.
Figure 2 is an example of the respiration rate estimation applied to a 300â€‰s long record.
2.5. Kalman Filter Framework and Data Fusion
The KF is an optimal state estimation method for a stochastic signal [29] that estimates the state of a discretetime controlled process, , with measurement data , where and are governed by the linear stochastic difference equations:
The random variables and are independent, white, and possess normal probability distributions, ~ N(0,Q) and ~N(0,R). The matrices A, B, andH are the state transition, control input, and measure matrices, Q being the state noise covariance, R the measurement noise covariance, and an optional control input to the state . Further details on the KF algorithm can be found elsewhere [29].
Previously we employed the KF to estimate the systolic, mean, and diastolic blood pressure derived from ABP and HR from the ECG [15, 16]. In order to more heavily weight estimates derived from cleaner data, the SQI is used to adjust the measurement noise covariance, R. When the SQI is low, should be trusted less, and hence we force R to be large. This is achieved by modifying at the k th timestep as follows:
where is the signal quality of the k th segment of data and may be replaced by various measures of the underlying signal quality, such as the purity index defined in (2), that is, . This nonlinear weighting function therefore tends to unity as the value of tends to unity (at which point the measurement noise covariance is no longer affected by the SQI), forcing the KF to trust the current measurement given the baseline measurement noise covariance matrix . At low values of , tends to infinity (but in practice is limited to a large value to deal with issues of convergence) and therefore forces the KF to trust current measurements less. This is the key factor in the modified KF framework; we allow the KF to make a varying estimate of the noise covariance using independent signal quality estimates based upon domain knowledge of the underlying ECG, and not the KF itself.
The SQI of each heart beat was calculated 5â€‰s around each beat. Secondbysecond ECG and PAT SQI were acquired by calculating the median values of these beats within a moving 20â€‰s window with 50% overlap. Then, the ECG and PAT features and SQI were used by the KF to obtain the optimal breathing rate estimation on a 5â€‰sby5â€‰s basis.
2.5.1. Kalman Filter Initialization and Operation
Following Tarassenko et al. [1] and Mason [17], we pick the simplest form of the KF and set the state to be a scalar. We assumed that the breathing rate at each moment is approximately equal to the breathing rate at the next moment. After neglecting the control input , (3) then reduces to . In order to initialize the KF, one must estimate Q, the state noise covariance matrix, and R, the measurement noise covariance. R was similarly initialized to unity, noting that it is immediately modified by the SQI to reflect our trust in the data. Q was empirically adjusted to have an initial value of (5 breath/minute). Values of lead to the KF trusting the initial state estimate too little and not adapting to the new initial observations. Values of lead to the KF trusting the new observations too much, and simply following the new values too closely. The filter can then be run online with only a few iterations for convergence. The Kalman residual is then given as for every newly available measurement. (Note that here a new measurement is a new estimate of the respiratory rate from one of the sensors as described in Sections 2.2 and 2.4.)
2.5.2. Merging of Multiple Kalman Filter Estimates
In order to calculate a single estimate of the respiratory rate, estimates from individual Kalmanfilters must be fused in a manner that takes into account the uncertainty associated with each estimate. In general, it is possible to fuse any number of independent Kalman filtered estimates, , using the technique of Mason [17] and Tarassenko et al. [1] such that the final estimate at the k th timestep is given by
where and are the independent estimate and the associated uncertainty for the s th sensor at the k th timestep, respectively. Here is taken to be the innovation or the residuals, , associated with the Kalman filter estimate. In a recent paper [30] we proposed a modification to this approach where the SQIscaled innovations are given by . In this way, when one channel, say , is corrupted by artifact and the corresponding parameter estimate () is miscalculated, the SQI () will be low and the sudden change of will make the residual error () large. The weighted innovation () will therefore be large and the weighting for (which would be for two channels) will be small. The estimation of will then rely more on than .
In theory, each of the estimates can be recorded at different (or uneven) sampling frequencies. Adjustments to the innovation update sequence can be made to adjust for the differing sampling frequencies and the inherent confidences in the different recording equipment. In general, the innovationbased weighting function can be modified so that [17]
where is a "trust" factor for the s th channel of data. For example, may be decremented to a value of 0.5 for data derived from the ECG or pulse oximeter, to account for the fact that the derived signal is often less accurate for respiration rate estimation than the impedance pneumogram. For the purpose of respiration rate estimation, however, was set to unity for all channels.
3. Results
We compared seven respiration algorithms, three of which were derived from the ECG (using RS amplitude, QRS area, and RSA), one from the PAT amplitude oscillations, and three KFbased fusion algorithms.

(i)
Fusion 0 algorithm used no (or a constant) signal quality adjustment. This was accomplished by setting =1 in (5) for all cases.

(ii)
Fusion 1 used the ECG SQI metric for ECGderived respiratory signals and the signal purity index for the PATderived data.

(iii)
Fusion 2 algorithm used the signal purity index (for all derived respiration signals).
In the case of the real data, we report the performance of all seven respiration algorithms, while for the simulation studies only the ECGbased algorithms are considered. Note that, fusion 0 algorithm is equivalent to the method ofâ€‰â€‰Mason [17] which weights the individual respiratory rate estimates proportional to the inverse of the square innovations.
3.1. Simulation Results
In order to evaluate the respiration rate estimation algorithm, we used the simulated ECG data with varying respiration rates and under different heart rates (60â€“100 beats/minute) and signaltonoise ratio scenarios (10, 20, and 40â€‰dB). The heart rate dependence was negligible; therefore the results that follow are the average performances over all heart rates. Note that all values reported in this work are in root mean square (RMS) breaths per minute (BPM).
Results of these simulations (see Figure 3) indicate that the RS method is the best estimator for breathing rates in the range of 16â€“24â€‰BPM, the RSA method is best for breathing rates in the range of 8â€“12â€‰BPM, while the QRS area method is best for rates in the range of 16â€“24â€‰BPM. At the lowest SNR, the KF fusion algorithm provides a good estimate for the rates of 8â€“24â€‰BPM.
Table 1 summarizes the overall performance of each algorithm individually as well as the fusion results using the purity SQI. A comparison across a wide range of SNRs indicated that the fusion results remain fairly consistent while performance of the individual algorithms may vary (median RMS errors of 6.7, 6.4, and 5.0â€‰BPM for SNR of 10, 20, and 40â€‰dB, resp.).
3.2. Overnight Polysomnography Results
Figure 4 is an example of the KFbased fusion algorithm using four different respiration rate signals. The bottom plot is the fusion result using the purity signal quality index, while the proceeding plot (one to the bottom panel) shows the results of using no signal quality (thus, using a fixed measurement noise covariance matrix). In the 140â€“280â€‰s region, where the quality of the estimation is poor, the signal quality/KFbased fusion method shows clear improvements in the estimated respiratory rate.
Table 2 presents results of the average (RMS) error for all 30 patients in the overnight polysomnographic database. Four single channel algorithms and three fusion algorithms were evaluated. The four single channel algorithms were RS amplitude EDR_RS, QRS area (EDR_G), and RSAbased methods (EDR_RS) of deriving respiratory waveforms from the ECG and the pulse amplitude modulation method of deriving respiratory waveform from the PAT signal (PDR). Respiratory rates and puritybased signal quality indices were extracted from these waveforms using the AR modeling method of Section 2.4 and signal quality metrics of Section 2.3.
Note that the three fusion algorithms consistently perform better than any one single algorithm. In particular, when the SQI is used, a greater improvement (lower RMS error) is seen. Interestingly, results from our real data are better than our simulations, even at high SNR, indicating that even though the PAT derived respiration is particularly poor, the inclusion of this signal still improves the overall performance. This is because the PAT signal was occasionally good quality at times when the ECG was noisy.
4. Discussion and Conclusions
In this work several respiratory rate estimation algorithms have been presented which estimate a respiratory rate from the respiratory waveforms derived from ECG or PAT. They have been divided into three categories:

(1)
EDR algorithms based on beat morphology, namely, those based on ECG wave amplitude or QRS area (EDR_G),

(2)
EDR algorithms based on HR information, that is, respiratory sinus arrhythmia (EDR_RSA),

(3)
Algorithms based on pulse or PAT amplitude variations (PDR).
The choice of a particular EDR algorithm depends on the application. In general, EDR algorithms based on beat morphology are more accurate than those based on HR information, particularly at high respiration rates (16â€‰BPM) since the modulation of ECG by respiration is sometimes too small or embedded in other parasympathetic interactions. At all other rates the KFbased fusion approach is superior.
Electrocardiogramderived respiration algorithms based on both beat morphology and HR may be appropriate when only a singlelead ECG is available and the respiration effect on that lead is not pronounced. Although not detailed here, the power spectra of the EDR signals based on morphology and HR can also be crosscorrelated to reduce spurious peaks and enhance the respiratory frequency [31]. However, the likelihood of having an EDR signal with pronounced respiration modulation is better when the signal is derived from multilead ECGs; crosscorrelation with the HR power spectrum may in those situations worsen the results due to poor respiratory HR modulation. This is particularly true during active wakefulness [31].
The median value of the fusion based estimation only diminishes by 1.7â€‰BPM from an SNR of 40â€‰dB down to an SNR of 10â€‰dB. The reason for such robust performance is not only due to the effectiveness of the fusion technique but also due to the good performance of the EDR_RS and EDR_RSA in the presence of noise. This is because they only rely on accurate detection of fiducial points on the ECG and therefore are not affected by the presence of noise in the other segments of the ECG (note that the Rpeak location has the largest SNR on the ECG trace and the Spoint detection can be fairly reliable given the location of the Rpeak). It should also be noted that the QRS area method's effectiveness in allowing respiration rate estimation is sensitive to the size of the integration window, which was not optimized in this study. Further improvements are therefore possible. Specifically, the QRS integration window could be shortened at higher heart rates (which is often correlated with higher respiration rates) to compensate for shortening of the QRS interval at these higher heart rates and elevated sympathetic tone.
Our results on real subjects' data recorded during sleep indicate that although the PAT is often noiser that the ECG and provides a poorer estimation of respiratory rate, if included in our KFbased fusion framework, it still adds value to the estimation process, particularly when an SQI is used. Furthermore, our respiration signal quality index, Î“, is superior to the use of the ECG and PATspecific SQI metric in providing a trust metric to automatically discount noisy data sources. In other words, testing the derived signal for information content provides a better trust index of the derived respiration rate than does an estimate of the quality of the underlying data from which the respiration signal is derived. This may be because a good quality waveform does not always carry respiration information and may therefore degrade the KFbased fusion algorithm if a postprocessing quality metric is not used. It may of course be productive to look at using both signal quality metrics.
Although our gold standard respiration rate is derived from nasal thermistor, it is unlikely that the nasal thermistor always provides a clean and representative signal, being susceptible to movement artifact at least. If the nasal thermistor provided a perfect evaluation of the respiration rate, we would expect our results to be even better, with lower error rates for all algorithms. However, we do not expect any of our individual algorithms to improve substantially in such a scenario, because the nasal thermistor is independent of the other measurement methods. The improved results of our KFbased fusion approach would not be invalidated in this case, and perhaps improved.
We can use the results of our investigation to inform a more intelligent combination of signals. By studying Tables 1 and 2 we see that the RSA and QRS areabased algorithms perform relatively weakly at higher respiration rates (16â€‰BPM), which suggests that we should weight the RS amplitude more strongly when the latter algorithm indicates high respiration rates. This can be incorporated by making the trust factor,Î»_{ k }, respiration ratedependent. The exact dependence could be calibrated for a particular population or recording environment if sufficient data were available.
There are still certain topics in the field of derivedrespiration which deserve further study. One is the robustness of the derivedrespiration algorithms in different physiological conditions, robustness to longterm beat morphology variations due to, for example, ischemia. The study of multimodal respiratory patterns should be considered when estimating the respiratory frequency from physiological signals by techniques like, for example, spectral coherence, particularly with wavelets to deal with nonstationarities. Weighting of the individual algorithms based upon respiration rate (through adjustment of Î»_{ k } in (7)) also requires further exploration given our simulation results. Also, Î»_{ k } may be adjusted to weight certain sensor measurements over others which are less trustworthy by nature. For instance, our results on the real data indicate that in general the PATderived respiration may not be as reliable as the ECGderived respiration for our population. This may not be true (or may be reversed) for other populations, such as neonates for example. Therefore, demographics may inform the values of Î»_{ k }.
Although our approach automatically rejects noisy sections of data, it may be further improved by using statistical tests which can reject spurious frequencies that could have appeared just by mere chance. This may be accomplished in a nonparametric and nonstationary manner using a surrogate analysis approach [32], for example.
The added computational burden of our fusion step (see (6)) is negligible; being essentially a weighted average it involves only a very few divides and adds. Generally one would expect the derived respiratory waveforms and respiratory rates to be already calculated. In fact, signal qualities are generally computed in most monitors although they are often not available to the standard user. Even in the case that a signal quality is not available, the puritybased signal quality method utilized in this work is calculated in time domain using a finitedifferencing approach (see [27]) and is therefore computationally very efficient. The majority of the computation is in the standard ECG analysis, which must be performed regardless.
The most important point to emphasize about our approach is that it does not require any a priori knowledge of the (changing SNR) of any input signal. Therefore our approach is robust in a wide sense, weighting the best estimators at any given epoch to provide a consistently superior estimate to any single given technique.
It should also be noted that the method presented in this paper is quite general and can be extended to any set of sources that provide a respiratoryrelated oscillatory data, such as the photoplethysmogram (processed in an almost identical manner as PAT), an airway thermistor, a flow meter, an impedance pneumogram, an accelerometer attached to a patient's chest, or an infrared camera pointing at the patient's mouth and nose. Moreover, this robust KFbased fusion framework is extensible to any set of independent (or nearly independent) observations, providing that a suitable signal quality parameter can be defined.
References
Tarassenko L, Townsend N, Clifford G, Mason L, Burton J, Price J: Medical signal processing using the software monitor. Proceedings of IEE Conference on Intelligent Sensor Processing, 2001, London, UK 3: 14.
Miyasaka K, Kondo Y, Suzuki T, Sakai H, Takata M: Toward better home respiratory monitoring: a comparison of impedance and inductance pneumography. Acta Paediatrica Japonica 1994, 36(3):307310.
Nobel JJ: Physiologic monitoring systems, acute care. Pediatric Emergency Care 1992, 8(4):235237. 10.1097/0000656519920800000016
Moody GB, Mark RG, Zocola A, Mantero S: Derivation of respiratory signals from multilead ECGs. Proceedings of IEEE Conference on Computers in Cardiology, September 1985, Linkoping, Sweden 12: 113116.
Gil E, Mendez MO, Villantieri O, Mateo J, Vergara JM, Bianchi AM, Laguna P: Heart rate variability during pulse photoplethysmography decreased amplitude fluctuations and its correlation with apneic episodes. Proceedings of IEEE Conference on Computers in Cardiology, September 2006, Valencia, Spain 33: 165168.
Yi WJ, Park KS: Derivation of respiration from ECG measured without subject's awareness using wavelet transform. Proceedings of the IEEE Engineering in Medicine and Biology, the 24th Annual Conference and the Fall Meeting of the Biomedical Engineering Society (BMES /EMBS '02), October 2002, Houston, Tex, USA 1: 130131.
BailÃ³n R, SÃ¶rnmo L, Laguna P: A robust method for ECGbased estimation of the respiratory frequency during stress testing. IEEE Transactions on Biomedical Engineering 2006, 53(7):12731285. 10.1109/TBME.2006.871888
Johansson A, Ã–berg PÃ…, Sedin G: Monitoring of heart and respiratory rates in newborn infants using a new photoplethysmographic technique. Journal of Clinical Monitoring and Computing 1999, 15(78):461467.
Larsen VH, Christensen PH, Oxhoj H, Brask T: Impedance pneumography for longterm monitoring of respiration during sleep in adult males. Clinical Physiology 1984, 4(4):333342. 10.1111/j.1475097X.1984.tb00808.x
Gil E, Vergara JM, Laguna P: Study of the relationship between pulse photoplethysmography amplitude decrease events and sleep apneas in children. Proceedings of the 28th Annual International Conference of the IEEE Engineering in Medicine and Biology Society (EMBC '06), 2006, New York, NY, USA 38873890.
O'Brien C, Heneghan C: A comparison of algorithms for estimation of a respiratory signal from the surface electrocardiogram. Computers in Biology and Medicine 2007, 37(3):305314. 10.1016/j.compbiomed.2006.02.002
de Chazal P, Reilly R, Heneghan C: Automatic sleep apnoea detection using measures of amplitude and heart rate variability from the electrocardiogram. Proceedings of the 16th International Conference on Pattern Recognition (ICPR '02), August 2002, Quebec, Canada 1: 775778.
Ishijima M: Monitoring of electrocardiograms in bed without utilizing body surface electrodes. IEEE Transactions on Biomedical Engineering 1993, 40(6):593594. 10.1109/10.237680
Park SB, Noh YS, Park SJ, Yoon HR: An improved algorithm for respiration signal extraction from electrocardiogram measured by conductive textile electrodes using instantaneous frequency estimation. Medical and Biological Engineering and Computing 2008, 46(2):147158. 10.1007/s115170070302y
Li Q, Mark RG, Clifford GD: Artificial arterial blood pressure artifact models and an evaluation of a robust blood pressure and heart rate estimator. BioMedical Engineering Online 2009., 8, article 13:
Li Q, Mark RG, Clifford GD: Robust heart rate estimation from multiple asynchronous noisy sources using signal quality indices and a Kalman filter. Physiological Measurement 2008, 29(1):1532. 10.1088/09673334/29/1/002
Mason L: Signal processing methods for noninvasive respiration monitoring, Ph.D. in Engineering Science. Department of Engineering Science, University of Oxford, Oxford, UK; 2002.
Mason CL, Tarassenko L: Quantitative assessment of respiratory derivation algorithms. Proceedings of the 23rd Annual International Conference of the IEEE Engineering in Medicine and Biology Society (EMBS '01), October 2001, Istanbul, Turkey 2: 19982001.
Clifford GD, Nemati S, Sameni R: An artificial vector model for generating abnormal electrocardiographic rhythms. Physiological Measurement 2010, 31(5):595609. 10.1088/09673334/31/5/001
AstrÃ¶m M, Santos EC, SÃ¶rnmo L, Laguna P, Wohlfart B: Vectorcardiographic loop alignment and the measurement of morphologic beattobeat variability in noisy signals. IEEE Transactions on Biomedical Engineering 2000, 47(4):497506. 10.1109/10.828149
Moody GB, Muldrow WK, Mark RG: A noise stress test for arrhythmia detectors. Proceedings of IEEE Conference on Computers in Cardiology, September 1984, Salt Lake City, Utah, USA 11: 381384.
Hamilton PS, Tompkins WJ: Quantitative investigation of QRS detection rules using the MIT/BIH arrhythmia database. IEEE Transactions on Biomedical Engineering 1986, 33(12):11571165.
Zong W, Moody GB, Jiang D: A robust opensource algorithm to detect onset and duration of QRS complexes. Proceedings of IEEE Conference on Computers in Cardiology, September 2003, Thessaloniki, Greece 30: 737740.
Zong W, Heldt T, Moody GB, Mark RG: An opensource algorithm to detect onset of arterial blood pressure pulses. Proceedings of IEEE Conference on Computers in Cardiology, September 2003, Thessaloniki, Greece 30: 259262.
Gil E, Vergara JM, Bianchi AM, Laguna P: Obstructive sleep apnea syndrome analysis in children by decreases in the amplitude fluctuations of pulse photoplethysmography: role of recording duration and heart rate variability. Proceedings of the 29th Annual International Conference of Engineering in Medicine and Biology Society (EMBC '07), August 2007, Lyon, France 60896092.
Gil E, MarÃa Vergara J, Laguna P: Detection of decreases in the amplitude fluctuation of pulse photoplethysmography signal as indication of obstructive sleep apnea syndrome in children. Biomedical Signal Processing and Control 2008, 3(3):267277. 10.1016/j.bspc.2007.12.002
SÃ¶rnmo L, Laguna P: Bioelectrical Signal Processing in Cardiac & Neurological Applications. Elsevier, Burlington, Mass, USA; 2005.
Akaike H: A new look at the statistical model identification. IEEE Transactions on Automatic Control 1974, 19(6):716723. 10.1109/TAC.1974.1100705
Brown RG: Introduction to Random Signal Analysis and Kalman Filtering. John Wiley & Sons, New York, NY, USA; 1983.
Clifford GD, Long WJ, Moody GB, Szolovits P: Robust parameter extraction for decision support using multimodal intensive care data. Philosophical Transactions of the Royal Society A 2009, 367(1887):411429. 10.1098/rsta.2008.0157
Thomas RJ, Mietus JE, Peng CK, Goldberger AL: An electrocardiogrambased technique to assess cardiopulmonary coupling during sleep. Sleep 2005, 28(9):11511161.
Nemati S, Abdala O, Bazin V, YimYeh S, Malhotra A, Clifford G: A nonparametric surrogatebased test of significance for Twave alternans detection. IEEE Transactions on BioMedical Engineering. In press
Acknowledgments
This work was supported by the Information and Communications University (ICU) in Daejeon, South Korea, the U.S. National Institute of Biomedical Imaging and Bioengineering (NIBIB), American Heart Association (AHA) (Grant 0840159N), and the National Institutes of Health (NIH) (through Grant nos. R01 EB001659, HL73146, HL085188â€“01A2, HL090897â€“01A2, K24 HL093218â€“01A1, and the training Grant T32â€“HL07901). The content of this document is solely the responsibility of the authors and does not necessarily represent the official views of the ICU, the NIBIB, or the NIH. The authors would also like to thank Professor Pablo Laguna and Eduardo Gil for their data sharing and many helpful discussions concerning signal quality.
Author information
Authors and Affiliations
Corresponding author
Rights and permissions
Open Access This article is distributed under the terms of the Creative Commons Attribution 2.0 International License ( https://creativecommons.org/licenses/by/2.0 ), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.
About this article
Cite this article
Nemati, S., Malhotra, A. & Clifford, G.D. Data Fusion for Improved Respiration Rate Estimation. EURASIP J. Adv. Signal Process. 2010, 926305 (2010). https://doi.org/10.1155/2010/926305
Received:
Revised:
Accepted:
Published:
DOI: https://doi.org/10.1155/2010/926305
Keywords
 Kalman Filter
 Respiratory Sinus Arrhythmia
 Breathing Rate
 Peripheral Arterial Tonometry
 Measurement Noise Covariance